During interrogation of a dual chamber MicroPort pacemaker (KORA 250 DR) you see the following:
Video Case | Arrhythmia
0 of 1 Questions completed
You have already completed the case before. Hence you can not start it again.
Case is loading…
You must sign in or sign up to start the case.
You must first complete the following:
0 of 1 Questions answered correctly
Time has elapsed
- Review / Skip
What is the correct sequence of events?
atrial events intermittently occur within the post ventricular pacing blanking
DDD mode with 2:1 tracking of atrial tachycardia
switch to DDI mode
first suspicious ventricular cycle since it includes at least one Ar
algorithm waits for sliding window of 32 ventricular cycles
the ventricular rate is progressively decreased to reach the Sensor Driven Rate or the Basic Rate